Hello everyone, normally in my workplace exchanges of programs between different cmm or measurement equipment are experienced and being honest I have not found the best method to document these practices as well as to establish the best limit of correlation between the measurement equipment, in last days I found that the statistical control tool has a socalled "Dispersion Analysis" and offers something called "Pearson Coefficient" which provides a data indicating the strength of the correlation between two variables, does anyone have a method to document the correlation between different measuring equipment?
What is the best correlation method?
Collapse
X

We used paired T tests, where we measure 36 sample parts on equipment A, assess average of the resultant data to the same 36 parts on equipment B. If the delta between means is less than 10% (of the product's dimensional tolerance), the equipment is considered equivalent.Last edited by louisd; 10252019, 04:02 PM.
 1 like
Comment

Here's Paired T explanation from Minitab help:
Use Paired t to do the following: Determine whether the mean of the differences between two paired samples differs from 0 (or a target value).
 Calculate a range of values that is likely to include the population mean of the differences.
The paired ttest is useful for analyzing the same set of items that were measured under two different conditions, differences in measurements made on the same subject before and after a treatment, or differences between two treatments given to the same subject.
For example, the manager of a fitness facility uses a paired ttest to determine whether a group of participants improved their fitness after an 8week program. If there is a difference in fitness measurements before and after the program, the manager can use the confidence interval to determine whether the difference is practically significant.
Comment

Originally posted by mckenzie View Postim not quite sure exactly what you're asking but, here goes, at first guess could you not run a gage r&r using the same part on the various cmm's/measurement equipment?
Comment

Originally posted by louisd View PostHere's Paired T explanation from Minitab help:
Comment

the theory is if you have already confirmed the measurement method is sufficient to pass a gage study (Crossed or Anova Gage R&R), and both pieces of equipment have comparable accuracy, there is no reason to conduct a Gage R&R study again on the similar piece of equipment. A statistical assessment of the measured values using Paired T, also looks at the standard deviation. if stddev is significantly higher for the new equipment, it would be a concern. otherwise, paried t comparing the means of each sample set will suffice.

Related Topics
Collapse

by SchrocknrollHow many of you are required to do a correlation study on your program before accepting final product? This is our procedure, but I question the reasonableness...

Channel: PCDMIS for CMMs
05162006, 03:50 PM 

by hpwanlessI'm having trouble correlating measurements from a micrometer to CMM. We used micrometers for the longest time to calibrate the machine with plastic...

Channel: PCDMIS for CMMs
08272009, 12:02 PM 

by MJBolsterI've been in the machining trade 25+ years and Inspection 3+. Working
with Pcdmis 1 yr. (actually, in college, I was an art major, go figure)...
Channel: PCDMIS for CMMs
07062006, 12:35 PM 

Many of you know I am a strong advocate of some sort of process control on CMMs.
I have put together some of the lessons learned over the...
Channel: PCDMIS for CMMs
03232006, 11:06 AM 

by ArenwHas anyone on here had to perform OEE studies for CMMs? I've been tasked with conducting an Operating Equipment Efficiency study on our CMMs and I'm...

Channel: PCDMIS for CMMs
09122012, 10:27 AM 
Comment